maybe you want to know these
Non-invasive ventilator technical parameters (YH-830)
Ⅰ.Scope of application:
For adult patients with sleep apnea, hypoventilation syndrome, for two-level non-invasive ventilation in the hospital or home.
Ⅱ. the main function:
1. Different treatment schemes can be provided according to patients’ needs, from CPAP to S, T, S / T, and VGPS.
2. Adjustable trigger sensitivity.
3. Real-time monitoring and recording of blood oxygen saturation.
4. Prompt function.
5. Record function (SD card).
6. Maximum leak compensation function.
7. Cloud platform management function for wireless uploading data (WIFI module).
Ⅲ.the key technical parameters:
1. Working mode: CPAP, S, T, ST, VGPS.
2. Ventilation parameters:
2.1 Inspiratory pressure: 4-30 cmH2O (0.5 cmH2O interval)
2.2 Expiratory pressure: 4-30 cmH2O (0.5 cmH2O interval)
2.3 Treatment pressure: 4-20 cmH2O (in CPAP mode, 0.5 cmH2O interval)
2.4 Delayed boost: 0-45 minutes (interval of 5 minutes)
2.5 Breathing frequency: 5-50 times / minute (interval 1 time / minute)
2.6 Breathing ratio: 10-70% (interval of 1%)
2.7 Rise time: 1-5 steps (1 step interval, 1 step rises fastest, 5 steps rises most slowly)
2.8 Inhalation sensitivity: 1-5 steps (1 step interval, 1st step triggers fastest, 5th step 1st step triggers slowest)
2.9 Exhalation sensitivity: 1-5 (1 interval, the 1st trigger is the fastest, 5th and 1st the slowest)
2.10 Tidal volume: 50-1500ml (effective in VGPS mode, interval 50ml)
3. VGPS (capacity guarantee pressure support) mode:
3.1 Maximum IPAP value adjustable range: 4-30 cmH2O
3.2 Adjustable range of minimum IPAP value: 4-30 cmH2O
3.3 EPAP value adjustable range: 4-30 cmH2O
3.4 Target Tidal Volume: 50-1500ml
Ⅳ. Display range of monitoring setting parameters:
4.1 Respiratory waveform can be displayed;
4.2 Current mode: CPAP, S, T, ST, VGPS
4.3 Inspiratory pressure: 4-30 cmH2O
4.4 Expiratory pressure: 4-30 cmH2O
4.5 Breathing frequency: 5-50 times / minute
4.6 Breathing ratio: 10-70%
4.7 Air leakage: 20-99.9lpm
4.8 Tidal volume: 50-1500ml
4.9 minute ventilation: 0-50lpm
4.10 Blood oxygen saturation: 70-100%
4.11 Pulse rate: 25-250bpm
4.12 Humidifier gear: 0-6 gear
4.13 Delay boost: 0-45 minutes
Ⅴ. Prompt items:
5.1 High-voltage sound prompt: Turn on / off sound prompt
5.2 Low minute ventilation sound prompt: Turn on / off sound prompt, can be set to 1-10L (interval 1L)
5.3 Air leakage sound prompt: on / off
5.4 Power off sound prompt: on / off
5.5 Choking voice prompt: On / off, choking time can be set to 10-40 seconds (interval of 5 seconds)
5.6 Low Tidal Volume Sound Prompt: On / Off, Set Tidal Volume Value Range 50-500ml ((Interval 50ml))
6.1 Weight: about 1500g
6.2 Operating temperature: 5-35 ° C
6.3 Relative humidity: 10% -90%
6.4 Power supply: input is AC 100-240V, 50-60Hz, 1.8A Max; output: DC 24V, 3.33A.
Just enter your email address and order number is ok.
The deep outer and shallow inner should know that the nose clip should be kept in mind.
Cover the mouth, nose and chin, and stretch the folds.
Finger press the metal strip and fit it all around.